|Typical Properties||Average Result|
|Extrusion Rate†, Part A||85 g/min|
|Extrusion Rate†, Part B||110 g/min|
|Work Time||70 hours|
|Cured: 5 minutes at 150 °C (302°F). Stabilize for 3 hours minimum at ambient temperature and humidity|
|Durometer, Type A||10|
|Tensile Strength||450 psi (3.1 MPa)|
|Tear Strength||65 ppi (11.5 kN/m)|
|Tissue Culture (Cytotoxicity Testing)||Pass|
|Elemental Analysis of Trace Metals||Pass|
These properties tested on a lot-to-lot basis. Please contact NuSil Technology for assistance and recommendations in establishing particular specifications.
†(Performed using a SEMCO® model 250-A pneumatic gun with a ⅛ nozzle orifice and 90± 5 psi air pressure)
Recommended Primer MED1-161
Note: Some bonding applications may require the use of a primer.
50 mL Side-by-Side Kit
200 mL Side-by-Side Kit
400 mL Side-by-Side Kit
2 Pint Kit (910 g)
2 Gallon Kit (7.28 kg)
10 Gallon Kit (36.4)
The warranty period provided by Polymer systems Technology ltd (hereinafter “PST”) is 12 months from the date of shipment when stored below 40°C in original unopened containers. Unless PST provides a specific written warranty of fitness for a particular use, PST’s sole warranty is that the product will meet PST’s then current specification. PST specifically disclaims all other expressed or implied warranties, including, but not limited to, warranties of merchantability and fitness for use. The exclusive remedy and PST’s sole liability for Packaging Warranty 50 mL Side-by-Side Kit 12 Months 200 mL Side-by-Side Kit 400 ml Side-by-Side Kit 2 Pint Kit (910 g) 2 Gallon Kit (7.28 kg) 10 Gallon Kit (36.4 kg), Inc. All rights reserved. Trademarks are owned by PST or its affiliates unless otherwise noted. breach of warranty is limited to refund of purchase price or replacement of any product shown to be other than as warranted. PST expressly disclaims any liability for incidental or consequential damages.
PST believes, to the best of its knowledge, that the information and data contained herein are accurate and reliable. The user is responsible to determine the material’s suitability and safety of use. NuSil Technology cannot know each application’s specific requirements and hereby notifies the user that it has not tested or determined this material’s suitability or safety for use in any application. The user is responsible to adequately test and determine the safety and suitability for their application and NuSil Technology makes no warranty concerning fitness for any use or purpose. PST has completed no testing to establish safety of use in any medical application. PST has tested this material only to determine if the product meets the applicable specifications. (Please contact PST for assistance and recommendations when establishing specifications.) When considering the use of PST products in a particular application, review the latest Material Safety Data Sheet and contact PST with any questions about product safety information. Do not use any chemical in a food, drug, cosmetic, or medical application or process until having determined the safety and legality of the use. The user is responsible to meet the requirements of the U.S. Food and Drug Administration (FDA) and any other regulatory agencies. Before handling any other materials mentioned in the text, the user is advised to obtain available product safety information and take the necessary steps to ensure safety of use.
PST disclaims any expressed or implied warranty against the infringement of any domestic or international patent/intellectual property right. PST does not warrant the use or sale of the products described herein will not infringe the claims of any domestic or international patent/intellectual property right covering the product itself, its use in combination with other products, or its use in the operation of any process.